|Frequently Asked Question 7|
|Home > F.A.Q. > Answers > Question 7|
Will the ozone analyzer OGA/C you suggested for measuring of ozone generator output produce the reading in pounds per day?
Sorry, if I sound a little ignorant, I am. BUT I'm the only one here who has a good grasp on English, so I get stuck with the duty.
=== We have a full line of ozone monitors for monitoring of ozone gas concentrations.
The OGA/C ozone analyzer in the suggested configuration (Base model + Enclosure with 4-digit LCD display and Ozone IN/OUT ports) will produce a digital reading in the form of [mg/l].
It is important to understand, that Ozone Concentration is measured. However, the Total Ozone Output is calculated. In the calculation you will employ the following variables:
....and you will be using the following formula:
Ozone Concentration [mg/l] x Ozone Flow [LPM] x 60 x 24 => [mg/day]
If you wish to convert [mg/day] to [Lb/day], then you have to take into consideration, that the conversion is rate is 1Lb = 454,000mg
12x 20x 60 x 24=> 345,600 [mg/day]
345600 / 454000 => 0.76 [Lb/day]
=== I hope this helps,
=== Take Care
=== The OzoneLab™ Team